Important note on calibration bugs in ver 0.0.0.15 of RecVox
The first release 0.0.0.15 of RecVox had the following bugs in the SPL calibration:
If it was performed with a calibrator, all levels were registered 3.7 dB too high.
If the calibration was performed with a loudspeaker or with the voice, levels were correct, but the overload limit was 3.7 dB too low.
The bugs are fixed from version 0.0.0.17.
Note: This software is actual freeware, with no ads, time limits, spyware or anything like that.
Note: Tolvan Data takes no responsibility whatsoever for the quality of the software, by using the software you take full responsibility for validating that the software meets your requirements, and whatever legal requirements that might be applicable in your country. In particular, Tolvan Data does not accept the role as a supplier of medical devices (sw. medicintekniska produkter).